Easy Operation PLC Touch Screen Stainless Steel Battery Burning And Ejection Test Equipment
IEC60086-4 clause 6.2.6,6.2.8 ,IEC62133,UL 1642,UL2054,GB31241 and etc.
This battery burning tester is necessary testing equipments for various battery manufacturer, quality inspection & control department. With features of easy operation, safety reliable, high accuracy, it becomes ideal testing machine for battery burning and ejection testings.
1.Fix the mesh on top of the burner with 38.1mm distance,each 25.4mm mesh with 20 holes, steel wire diameter 0.43mm.
2. Ignition the burner, simultaneously observe the battery burning situation,till battery explode or burn out. And time the whole burning procedure.
|Burner||Bunsen burner,pipe inner diameter 0.375inch(9.5mm)length about 100mm|
|Flame Applicate Time||0~999.9s±0.1s|
|Flame Continuous Time||0~999.9s±0.1s|
|Testing Bar Diameter||1/4 inch(6.35mm)×14 inch(355.6mm)|
|Test Cover Width||24 inch(610mm)×14 inch(355.6mm),Octagonal|
|Test Cover Height||12inch (305mm)|
|Testing Aluminum Mesh Diameter||0.010 inch (0.25mm) 16hole×16hole|
|Octagonal stainless steel net cover||Distance between each other 2 feet (610mm),Octagonal net cover height1 feet (305mm)|
|Testing Hole Diameter||102mm|
|Fire Ignition Device||Auto fire ignition|
|Display Method||PLC Touch Screen|
|Control Method||Remote control, whole test procedure fully automatic|
|Burning Gas Requirement||High purity liquefied petroleum gas(Customer Prepare)|
|Bottom||Installed with universal wheel|
|Observation Window||390 X 360mm(20mm thickness tempered explosion proof glass)|
|Timer||4 bytes display ,0~9999(Hour,Minute,Second switchable)|
|Inner chamber Material||SUS304 Stainless steel|
|Outer Chamber Material||Cold rolled powder painted|
|Smoke Extractor||Φ 150mm on back side|
|Overall Dimension||W940XD780XH1620 mm|
|Power Source||AC 220V 50HZ|
1. Battery burning ejection test & particle test is high dangerous destructive test,testing personal must be well trained with related safety & emergency handling knowledge, avoid personal injuries.
2.Burning device should put in the filtered air ventilate chamber,pay attention the waste air exhaust should comply with related national standards.
3. Testing personal must wear smoke proof expiratory device and filter expiratory devices, avoid suction harmful gas.
4. Before burning test,should strictly check the air pipe & connections sealing performance.
5. Multi fire extinguish devices should prepared close to the testing machine.
6. During testing or the bunsen light not totally cooling down, forbid to touch the sample burner or mesh.
7. After finishing test, should lock all the valve and well do the machine usage records, avoid air leakage of the gas tank.
8. Regularly check if the gas pipe broken or not,if broken, do stop use it,and replace with new ones.
|Stainless Steel Mesh||2piese|