3mm Steel Tumbling Single Barrel 500mm Height With 5rpm Speed Test Machine
This tester conforms to the standard requirement of VDE0620 ,IEC60068-2-32,BS1363.1,EN50075,UL60730. it is used to check the drop resistance of electrical accessories. to assess the effects on a specimen of simple standard tests intended to be representative of the fall likely to be experienced during rough handling, or to demonstrate a minimum degree of robustness, for the purpose of assessing safety requirements.
Standard: IEC 60884-1 2013 clause 24.2, IEC 60598-1, IEC 60998.1, VDE0620, IEC 60068-2-31, BS1363.1, EN50075, UL60730, IEC 60312-1 and so on
Application: This device is suitable for test the drop resistance of plugs and socket-outlets, surface type mounting boxes, screw glands, gland covers, switch and other electrical accessories.
Test sample: plugs and socket-outlets, surface type mounting boxes, screw glands, gland covers, switch and other electrical accessories.
Feature: This tester can preset test times, automatic stop when set test times completed.
|Drop rate||10time per min|
|Barrel number||Single barrel|
|Application||Plug type lamp ballast / transformer and power socket install luminaries, plug device|
|Standards||IEC60598-1 figure25, VDE0620-1 figure27|
|Power supply||220V 50Hz|
1. Set the number of drop, it will immediate effect after reset. User needs to press the reset button when change setting value of the drop times each time, otherwise, the counter will according to the original setting value to work.
2. Need to close the door after the specimen is put inside.
3.Press the "Start" button and start the test.
4.The test machine will stop working when achieve to the setting value, press the “Reset” button to eliminate the sound alarm.
5. Take out the sample, and judge the test results.
Note: The power supply of this machine is single phase 220V±10%, 50HZ, the voltage should be guaranteed within the specified range, otherwise it will affect the rotating speed of the barrel (and will damage the motor).
When the box body is stopped in the inclined position, re-start the testing machine, make it stop when the box goes to the proper position, so that not to damage the machine transmission parts.